The following procedure is recommended:
Install the AutoSmartIRt/c as close as practical to view the target material to be measured.
Wire the AutoSmartIRt/c to the controller, PLC, etc. in standard fashion (see PINOUT and WIRING DIAGRAM)
Bring the process up to normal operating temperature and measure the actual temperature of the target material with the Microscanner D-Series, Infrared Thermometer or any other reliable reference.
If using a PID controller: adjust “input offset,” “zero”, “low cal,” on the readout device to match the reading of your reference.
Installation Complete. (For OEM installations preset the same adjustments. Individual calibration is not required.)